No Fix No Fee
- Reference Number – RW17181
- Manufacturer – Adjustamatic
- Equipment Type – Wired Adjustable Chair Control
- Date Of Arrival – 24th July 2020
- Progress – Payment Received
- Date of Return – 29th July 2020
Fault Description – Handset for an Adjustamatic adjustable bed. Stopped work. Probably loose lead connection.
Remedy – This Adjustamatic control arrived safely and on inspection the grommet for the cable was missing and tape was wrapped around the cable. On opening the plug to the PCB was detached meaning the cable was not reaching the remote itself. The plug was reattached and a replacement grommet fitted to prevent it from detaching again.